The HI98108 pHep+® Hanna ph meter tester provides high accuracy measurements in a rugged pocket-sized design. This meter features an exposed temperature sensor that will provide rapid automatic temperature compensated measurements. The simple two-button operation provides one button for ON/OFF, the other for calibration. Being lightweight and only 0.7″ thick, the pHep+ is extremely ergonomic and fits comfortably in your hand.
Compact & waterproof design
Cloth junction provides a larger surface area than a standard ceramic junction
Great for environmental, field, and laboratory testing
The HI98108 tester features a large multi level LCD which displays both pH and temperature readings simultaneously. The pH readings are displayed with a 0.01 pH resolution and with an accuracy of ±0.10 pH. The pH range of the HI98108 is from 0.00 to 14.00 pH. An exposed temperature sensor allows for rapid automatic temperature compensated pH measurements. Temperature can be set to display in ℃ or ℉. The LCD screen has stability and calibration tag indicators. The battery percent level is displayed at start up alerting the user to the remaining battery power that is available.
The pH electrode uses a unique reference cell design that utilizes an extractable cloth junction. A clogged junction will typically exhibit erratic readings that will not stabilize. Simply extracting a small portion of cloth from the meter will clear any clog by exposing a fresh, clean surface. The ability to clear any clogging will extend the life of the pH electrode.
Hanna ph meter tester Calibration is performed by pressing the dedicated calibration button. Once in calibration mode the pH electrode can be calibrated to 1 or 2 points using the pre-programmed standard buffers (pH 4.01, 7.01,10.01). When the standard value is recognized and stability is reached, the meter automatically accepts and stores the calibration value.
Setup options include automatic shut off timing and temperature unit displayed. Automatic shut-off options are to turn the meter off after 8 minutes, 60 minutes, or disabled. To enter the setup mode the user presses a button located in the battery compartment.